# Basic Concepts: Surveyor Instruments

The instruments generally used in topography are the total station and the site level.

-the total station allows to measure horizontal angles, vertical and distances (the theodolite him, only measures horizontal and vertical angles).

-The level of construction allows to calculate differences of altitudes.

The setting up consists of placing the apparatus on the one hand vertically of the station and on the other hand in a horizontal plane.

To achieve this, the approach is as follows:

After opening and unfolding the tripod, install it above the station. A simple principle is to intersect the ground projections of the tripod's three branches on the station point.

The camera is fixed on the tripod tray. It must now be centered vertically in a horizontal plane.

The tacheometer and the tripod are firmly attached, the legs of the tripod are stable and tight, the camera is perfectly vertical to the station and the spherical level is adjusted. It is now a question of finishing this setting thanks to the toric level. (all instructions on the document)

When the unit is fully stationed, we can proceed with the survey of the land.

note the horizontal angle of another "stable" reference, distance ...

BN: For the sake of argument, it is estimated in topography that an angle difference of 1 mgr corresponds to 100 m to an error of 1.6 cm.

The inclined distance is that measured on the ground (along the distance-meter / reflector axis).

The reduced distance to the horizontal is the one actually separating two points: it is therefore the one that is plotted on the plane. V = vertical angle

Di = inclined distance

DH = reduced distance to the horizontal

DH = Di sin V

The site level is used to measure heights on a graduated rule (the chart) and by calculation to determine differences in altitudes between points.

1. Setting up a site level

Stationing involves placing the unit in a horizontal plane. same notion of total station

2. The sights

The sights are by definition horizontal and the readings are made on a test pattern by reading the graduation of the horizontal and main reticular wire.

3. Determination of differences in altitude

The determination of the differences of altitudes is calculated by a simple subtraction between the reading on the reference (Lref) and the reading on the point to be measured (Lp).

-the total station allows to measure horizontal angles, vertical and distances (the theodolite him, only measures horizontal and vertical angles).

-The level of construction allows to calculate differences of altitudes.

**The total stations****1. Setup**

To achieve this, the approach is as follows:

**1.1 The tripod**After opening and unfolding the tripod, install it above the station. A simple principle is to intersect the ground projections of the tripod's three branches on the station point.

**1. 2 Coarse calibration with spherical level**The camera is fixed on the tripod tray. It must now be centered vertically in a horizontal plane.

**1.3 Fine adjustment with the toric level**The tacheometer and the tripod are firmly attached, the legs of the tripod are stable and tight, the camera is perfectly vertical to the station and the spherical level is adjusted. It is now a question of finishing this setting thanks to the toric level. (all instructions on the document)

**2. sights and surveys****2.1. directions**When the unit is fully stationed, we can proceed with the survey of the land.

note the horizontal angle of another "stable" reference, distance ...

BN: For the sake of argument, it is estimated in topography that an angle difference of 1 mgr corresponds to 100 m to an error of 1.6 cm.

**2.2. Aimed at a prism with a total stations****3. Slope distance and reduced distance to the horizontal**The inclined distance is that measured on the ground (along the distance-meter / reflector axis).

The reduced distance to the horizontal is the one actually separating two points: it is therefore the one that is plotted on the plane. V = vertical angle

Di = inclined distance

DH = reduced distance to the horizontal

DH = Di sin V

**4. Altitude of a point** V = angle vertical Di = distance inclinĆ©e DH = distance horizontale Ht = hauteur de l’instrument ∆H = diffĆ©rence de hauteur de la visĆ©e Hp = hauteur du prisme Zs = altitude connue de la station Zp = altitude du point visĆ© (Ć dĆ©terminer) ∆H = diffĆ©rence de hauteur de la visĆ©e Zs = altitude connue de la station | ∆H = Di cos V Zp = Zs + Ht + ∆H – Hp Zp = Zs + Ht + Di cos V –Hp Zp = altitude du point visĆ© (Ć dĆ©terminer) |

**The level of construction**The site level is used to measure heights on a graduated rule (the chart) and by calculation to determine differences in altitudes between points.

1. Setting up a site level

Stationing involves placing the unit in a horizontal plane. same notion of total station

2. The sights

The sights are by definition horizontal and the readings are made on a test pattern by reading the graduation of the horizontal and main reticular wire.

3. Determination of differences in altitude

The determination of the differences of altitudes is calculated by a simple subtraction between the reading on the reference (Lref) and the reading on the point to be measured (Lp).